PARKS FLO-LAB 2100-SX
ISO 500 D power supply (UL544 Spec.):
120 V ac, ~, 60 Hz. 3 amperes typical, 5 amperes peak @ 120 V AC, 500 VA.
The POWER cord must be plugged into a 'HOSPITAL ONLY' or 'HOSPITAL GRADE' alternating current power outlet.
IR Remote (model 18-B) Batteries:
Three standard "AAA" alkaline batteries.
ENVIRONMENTAL CONDITIONS FOR TRANSPORT AND STORAGE
Ambient temperature: Range, -40 deg F to 158 deg F (-40 deg C to +70 deg C).
Relative humidity: 10% to 100%, condensing.
Atmospheric pressure: Range, 500 hPa to 1060 hPa.
IPXO rating: Degree of protection against ingress of water ..... none provided.
Temperature range: 50 deg F to 104 deg F (10 deg C to 40 deg C).
MAINTENANCE & CLEANING
The Flo-Lab operator or Biomed person should see "MAINTENACE" in the APPENDICES section of the manual.
Repair and Maintenance Literature is also located in PARKS 'Bioengineering Manual'. Circuit diagrams, component part lists, descriptions, calibration instructions, and other information are supplied to assist qualified technical personnel to repair parts of equipment which are designated by the manufacturer as repairable.
Calibrate every 1000 hours or once a year.
See PARKS 'Bioengineering Manual' for Cuff Volume Verification/Calibration.
For information on setting up, operating and servicing the computer and printer supplied with the Flo-Lab, see the factory 'Owners Manual' or 'Users Guide'.
Turn off power and unplug instrument from wall outlet before cleaning. Loose dust accumulated on the outside of the instrument can be removed with a soft cloth or small paint brush. Dirt which remains can be removed with a soft cloth dampened in a mild solution of detergent and water. Abrasive cleaners should not be used.
High frequency (nominal 8 Mhz).
Low frequency (nominal 4 Mhz).
Optional skinny pencil probe (nominal 8 Mhz).
The exact frequencies of the instrument are either engraved
on the probe or are indicated by a label attached to the probe
cable. Be sure to reorder replacement probes by these frequencies.
User selectable for 28, 14, 7, 3.5 (Hz) or mean flow.
User selectable for normal or inverted recording.
Standard low-impedance stereo headphones.
Pneumoplethysmograph (volume pulse recording).
Sensor, Parks part # 832-8000-00.
AC-coupled (arterial mode pulsations only).
DC-coupled (venous mode gross volume changes).
PPG - None
VPR - User-selectable calibrated pulse volume measurement. Pressure +/- 2 mm @ 100 mm Hg, +/- 5 mm above 100 mm Hg.
Momentary (BP mode) or fill to preset (VPR mode).
BP mode - Linear, variable speed, user selectable 5-30 mm Hg/sec.
VPR mode - fixed 30 mm Gh/sec.
BP Bleed rate:
User-selectable 1-10 mm Hg/sec.
Servo controlled & solenoid activated.
18 button infrared remote for volume, position, size, mute, trace auto scale,
cuff inflator and deflator, freeze, save, escape, enter and four direction cursor.
Height: 55 inches, 140 cm.
Width: 26 1/2 inches, 67 cm.
Depth: 30 inches, 76 cm.
Weight: 117 pounds, 60.66 kg.
Specifications subject to change without notice.
For Flo-Lab product information, please refer here.